Corometrics Aloka SSD 620
• Black and white multimodality ultrasound
• Single or dual selectable display mode
• 9-inch diagonal, horizontal scanning high
resolution TV monitor
• Applications: Ob/Gyn, general and
Scanning Method Electronic linear scan. Electronic convex sector scan
Display Mode B-mode (single or dual selectable), B/M – mode, M-mode
Linear Section Main frequency: 2.5 – 7.5 MHz, depends on probe. Scanning width: maximum
12.5 cm (depends on probe). Display depth: maximum 22 cm (depends on
probe). Focusing method: electronic dynamic focusing + acoustic lens. Image
direction: laterally (left/right) and longitudinally (up/down) changeable. Frame
rate: Maximum frequency 3.5 MHz. Probe: effective field of view 96 mm
Convex Sector Section Main frequency: 3.5-7.5 MHz, depends on probe. Scanning angle: maximum
90° depending on probe. Focusing method: electronic dynamic focusing +
acoustic lens. Image direction: laterally (left/right) and longitudinally (up/down)
changeable. Frame rate: maximum 30 frames/second. Standard convex: main
frequency 3.5 MHz. Sector probe: scan angle 62.5°
Viewing Monitor 9” diagonal, horizontal scanning high-resolution TV monitor.
Sensitivity Control GAIN: continuously variable (common to B- and M-modes). STC: depthcorresponding
8-step slide control.
Contrast Selectable 8 steps
Image Processing AGC (for B and M modes). Gamma curve: three kinds for optional
photographing unit. Polarity reversal. Post process: rejection
Automatic Displays Date and time; probe data (frequency); gain; acoustic power; focusing position;
measurement data; etc.
Measuring Functions Distance; area; circumference; velocity; time interval; heart rate; volume.
Calculating Functions Ratio; gestational age calculation; fetal weight calculation; left ventricule
Power Requirement 115/120/220/240 V ± 10%, approximately 210 VA
Dimensions 430 mm W x 800 mm D x 1,200 mm H
Weight Approximately 80 kg